Manual mode lets you position two cursors either vertical or horizontal , and the positions and difference are displayed. The zoom feature of the Rigol DSE oscilloscope. Home Questions Tags Users Unanswered. There is also this handy driver-less screenshot software that works over the Ethernet connection. Make sure you're controlling the trace you think you're controlling. If you use manual remember to set subnet mask and gateway so that packets will be routed. Now the hard part begins.
|Date Added:||7 November 2004|
|File Size:||69.30 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Welcome to Yes Thomas!
How do we handle problem users? Also make sure you're triggering from the right channel - the oscilloscope likes to default to using Channel 2 as the trigger. Controlling the channels If you've used an oscilloscope with separate controls for each channel, you may expect the knobs near CH1 to control channel 1, and the knobs near CH2 to utrascope channel 2.
Label oscilloscope probes with colored rings that match the trace colors. Thank you so much once again! If you have any additional oscilloscope tips, please leave a comment. Is this correct or am I possibly missing something? To set the user interface language, press "utility" then the the fourth soft button, then rotate the menu knob, then press it. An alternative is to run the slightly clunky UltraScope software on your computer, which gives you access to the oscilloscope via USB. However, there's also a trigger menu button under the trigger knob and a time base menu under the horizontal position knob.
There is also this handy driver-less screenshot software that works over the Ethernet connection.
RIGOL Oscilloscope DSE Ultrascope Software Install Problem
It's much easier to rename the file on your computer than on the oscilloscope. This is my first 'scope. Asked 1 year, 6 months ago. I am just trying to see riggol it is possible to bring both X and Y cursors onto my screen at the same time, to allot for measure capacitors.
You mention a dedicated "print" button - what button are you describing. Newer Post Older Post Home. For the price I love it. Improving the question-asking experience.
Pushing Ultdascope Level sets it to zero.
In mid-range settings, however, the pan range is fairly limited about a factor of 5 compared to how deep you can zoom with sofware horizontal scale knob about a factor of with Long Memory. If you use manual remember to set subnet mask and gateway so that packets will be routed.
There are three choices documented on page of the Users Guide. The Measure mode provides lots of automated measurements, but often doesn't measure what you want.
There's additional documentation here. Instead this button sets the controls to seemingly-random values to aotomatically display your traces. It did not come with the Logic Analyser cable. Don't forget display persistence mode.
I have done some digging with Wireshark and it appears the answer to my question is it can't be done. It works fine on the local LAN and apparently uses port Sign up or log in Sign up using Google.
Subscribe to RSS
Can I make up this cable or is it better to buy one? Please ignore the first portion of my post, which mentions a video.
Email Required, but never shown. Sign up using Email and Password.